With the development of skills, manufacturers have given more and more new functions to surgical shadowless lamps, so that surgical shadowless lamps can better provide help for medical workers. Today, we take the new shadowless lamp as an example to talk about the specific operation steps of its mutual control function.
Surgical shadowless lamp mutual control function operation steps mutual control version / with camera mutual control version - this lamp application method: (The following takes the main lamp as an example, and the secondary lamp application method is the same.) The meaning of the use of this lamp: the main light controller controls the main light, and the secondary light controller controls the secondary light). Lighting form selection operation: as shown in Figure 1-1, the screen lights up after power-on, click the main light button to enter the general version interface and click the power button to turn on the light.
Click the Specification Form button to enter the Specification Lighting Form; Click the low-light form button to enter the low-light lighting form; Click the laparoscopic form button to enter the laparoscopic surgical lighting form, if the spot electric conditioning function and the red and green light compensation function are turned on when setting, the screen interface after power-on is shown in Figure 1-2. Click the Specification Format button to enter the Specification Lighting Form.
Click the canonical form button again to enter the low-light lighting form; A third click takes you to laparoscopic surgical illumination. Click the red and green light compensation button to open the red and green light form, repeat click to adjust the red and green light brightness, brightness adjustment operation: in the above three forms, determine the power button is open, click the brightness plus or minus button to adjust the brightness of the shadowless lighting.