Electric operating beds have been popularized in all townships and towns, which are commonly used in hospitals. Doctors often need the electric operating table when performing surgery on patients. We should do a good job on daily maintaining electric operating beds,
Do you know what problems electric operating tables often present? How do I repair it? Today, I have a small editor to introduce you in detail:
If only the forward tilting action is not, and all other actions are, then it is clear that the tightening pump operation is normal, only the corresponding contact switch or relative solenoid valve should be faulty. Judging the quality of solenoid valves is generally determined by 2 methods. One is to measure the resistance with a multimeter, and the other is to put it on metal to see if there is suction.
Of course, if the solenoid valve closing action is normal. The above problems can also occur when the oil circuit is blocked. If there is not only no forward movement, but also other movements, then there is a crunch pump problem. First check should be on pump.
Check whether there is a voltage on the crunch pump, and whether the resistance of the crunch pump is normal or not. If the above is normal, it is generally formed by the failure of the commutator capacitor.
The manufacturer of electric operating table will introduce you here in detail today. If you have any more questions, please feel free to call us. We are dedicated to your service.