Electric operating tables in hospitals or private hospitals play an important role in the operating room, which can facilitate doctors in the operation, but if it fails, it will affect the use, and then we will introduce the reasons for the failure of the electric operating bed control panel:
1. The electric operating table has no action, that is, the ten action keys on the control panel do not respond, which may be caused by no AC 220 volt voltage input: the wall socket has no electricity or the 5 amp fuse is burned.
2. The failure of the five actions in the left or right column on the control panel may be caused by damage to one of the two TP137 pipes below the control panel circuit board.
3. The yellow line or orange line in the connecting line under the control panel is crushed in the middle: the bad coil of the electromagnetic reversing valve will also cause problems in the control panel.
4. The ten action keys on the control panel are inconsistent with the actual bed action, which is generally due to the long-term immersion of the silicone broad-core sealing gasket in the electromagnetic cabinet that controls the corresponding action. As a result, the solenoid valve cannot be closed when it is not working.
If you find the above problems, you can follow the flowchart. Use the backward push method to find out the location of the fault step by step, repair it in time, and pay attention to cleaning and maintenance at ordinary times. As a professional manufacturer of operating room equipment, we will introduce it to you in detail. If there are more enquiries related to the operating table series, you can call us at any time and we will give you a detailed answer.